BRAND offers the ideal pipette for every hand: BRAND’s new Transferpette® S. Transferpette® S models are are the perfect manual pipettes for demand-. Transferpette® electronic. F I R S T C L A S S · B R A N D. Gebrauchsanleitung. Operating Manual. Mode d’emploi. Instrucciones de manejo. 3. -8/ . Testing Instructions (SOP) testing of the piston operated pipette Transferpette®. BRAND additionally provides a calibration service (see chapter 7).
|Published (Last):||3 March 2006|
|PDF File Size:||3.62 Mb|
|ePub File Size:||13.94 Mb|
|Price:||Free* [*Free Regsitration Required]|
Limitations of Use The instrument is intended for the pipetting of liquids within the following limitations: Transferpete shaft Tip cone Fig. Laboratory Equipment Brand dispensette Operating Manual pages. Limitations of Use Purpose The vrand is an air-displacement system for pipetting aqueous solutions with medium density and low to medium viscosity. Consult the manufacturer for use in temperatures outside of this range.
Enter text from picture: Inspect the piston and O-Ring-seal for contamination. Optimum analysis results can only be obtained with quality tips. Autoclave the complete pipette without any further disassembling. Page 8 Pipetting c Let the pipetting button slide back slowly.
Ask your supplier or manufac- turer for the form. This instrument may sometimes be used with hazardous materials, opera- tions, and equipment.
Comments to this Manuals Your Name. Adjustment – Easy Calibration Adjustment — Easy Calibration The instrument is permanently adjusted for aqueous solutions.
BRAND TRANSFERPETTE S OPERATING MANUAL Pdf Download.
Pipetting Pipetting 5 ml and 10 ml instruments should only be used with the PE filter installed see page Hold the instrument vertically, with the sample in the tip for approximately 10 s. ISO prescribes rinsing the pipette tip once with the sample liquid prior to the actual pipetting process.
Got it, continue to print. Table of Contents Add to my manuals Add.
Safety Instructions Safety Instructions Please read the following carefully! The form can also be downloaded from www.
Discharge sample a Place the pipette tip against the wall of the vessel. Page of 25 Go. Pipette tips are disposables items! It is the responsibility of the user of this pipette to consult and establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Limitations Bgand Use Limitations of Use Purpose The pipette is an air-displacement system for pipetting aqueous solutions with medium density and low to medium viscosity. Pipetting 5 ml and 10 ml instruments should only be used with the PE filter installed see page Fitting the tip Use the correct tip according to the volume range or the color code.
Don’t lay the instrument horizontal when the tip is filled.
Parts will differ slightly depending on nominal volume of instrument. Warranty We shall not be liable for the consequences of improper handling, use, servicing, operating or unauthorized repairs of the instrument or the consequences of normal wear and tear especially of brxnd parts such as pistons, seals, valves and the breakage of glass as well as the failure to follow the instructions of the operating manual.
BRAND Transferpette S Operating Manual
Warranty Information Warranty We shall not be liable for the consequences of improper handling, use, transferpetts, operating or unauthorized repairs of the instrument or the consequences of normal wear and tear especially of wearing parts such as pistons, seals, valves and the breakage of glass as well as the failure to follow the instructions of the operating manual. Page 9 Pipetting Ejecting the tip Hold the pipette shaft over a suitable disposal container and press the tip ejection key to the stop.
Just send in the instruments to be calibrated, accompanied by an indication transferpetfe which kind of calibration you wish. Don’t show me this message again. Safety Instructions Please read the following carefully! To print the manual completely, please, download it. Eject the pipette tip. Checking The Volume Checking the Volume Depending on use, we recommend inspection of the instrument every 3 to 12 months.
For further information, refer to the accuracy table on pages 37 and The cycle can, however, be adjusted to individual requirements. Set nominal volume Set volume to the maximum volume indicated on the instrument see page 32 for procedure. Checking the Volume Depending on use, we recommend inspection of the instrument every 3 to 12 months.
Adjustment — Easy Calibration The instrument is permanently adjusted for aqueous solutions. Ensure that the tip is securely seated.